Failing that, I can’t think of much else – I wouldn’t recommend trying to fix a medical device, other than simple mechanical issues. Thank you very much. I thought it was the end of the line and I’ve had this thermometer since around 2004 and it just started to fail. Got my eight-year-old thermometer working again. These work OK and they fitted my Braun 6520 ThermoScan perfectly and performed as well as the sample box provided. I read recently that vaccines (e.g. The light will remain solid for 3 seconds to indicate that a successful temperature reading has been achieved. Hi Kevin, im having the same battery problem as yours in the same model. You’ll need tweezers to hold the wire while soldering. It was just dead. I’ll be removing the batteries when storing it from now on. Therefore, always take the . Is there a fix for that? I just used your article to fix my thermometer. The light will remain solid for 3 seconds to indicate that a successful temperature reading has been achieved. Thank you very much for sharing this guide and photos. Never submerge this thermometer in water or any other liquid. While there may be components that could be replaced with good soldering skills and suitable replacement parts, my opinion is that it’s not worth the risk of false readings where someone might be running a high temperature. Also, holding the power button for 10 or 20 seconds is a common way for designers to add a hard reset to electronic devices. I received these when I ordered them with my Braun Thermoscan 3, IRT 3030 model. If there’s no gold square left at all, the you’d have to make an equivalent, perhaps by coiling one end of the wire strand(s) and tinning with solder, before joining the other end to the copper trace you scraped back. We rarely change the cover, since it’s only used in our household – an alcohol wipe can be used to clean it. Kindly advise and thank you . Any other thoughts ?? If I had more patience I would’ve used baking soda and water solution with a q tip and/or steel wool or fine sand paper. The New Welch Allyn Braun ThermoScan PRO 6000 ear thermometer featuring advanced measurement technology, has the speed and simplicity you need to capture accurate and reliable temperature readings in a … Keep the cheapos for everything else! I think my thermometer have the corrosion problem too. I have this thermoscan and one of those gold square is broken. seasonal flu) are sold to the highest bidders first, rather than by triage, so it’s not just unscrupulous Amazon sellers that are at it. I’ve updated the info, thanks. Overview 2. It is easy to determine normal temperature range using the Braun ThermoScan. • Do not expose the thermometer to temperature extremes below - 4 °F or over 122 °F ( - 20 / 50 °C) or excessive humidity ( > 95 % RH non-condensing). This image above also shows the contact points on the circuit board for the battery connection. Anybody know how to open the battery compartment of Braun type 6013 thermometer? ExacTemp TM light will pulse while temperature is in progress. To help ensure accurate temperature measurements, the sensor itself is warmed to a temperature close to that of the human body. I don’t know much about CC licensing, but anything I donate my free time to has to remain unconditionally free. We have a Braun Thermoscan infra-red (IR) thermometer that has been working perfectly for about five years. We have found that the newer model is a bit more reliable when the tip is pointing snugly into the ear canal, but the glare from the screen still makes it difficult to take temperatures at night, when kids are drowsy or sleeping. However it’s pricey (very! Otherwise, you’d need to remove the PCB. If i could, what should i do ? It worked completely as you have described. Followed your description and fixed my thermo. Really pleased to hear that it helped. We fixed ours. I’m battling to get a screw driver to open it up but will try tom at hardware . Thanks so much. I have been searching for solution. Thank you very much for the help Kevin! If any feel sticky, or don’t seem to noticeably click, try using a tiny drop of WD-40 (or similar) solvent and wiggle the button with a pin to see if it frees up much. Mine got some corrosion on the left plate. 1,003 Problems and Solutions thermoscan. Our thermoscan had been working well, but I am not 100% sure if it is me or just in daylight (I tried using in a dark room). The usual fix is to gently scrape away the coating on the circuit board trace that connected to the gold square to reveal the copper. We have little kids at home and cannot live without a reliable ear thermometer and they are not cheap. Quick Overview. My approach would be to lightly tin what remains of the tab, and tin some multi-core wire. Our Braun is nearly 10 years old already and stopped working with no warning. I’m cautiously optimistic that the youngest generations are more savvy to resource squandering that our generation of pathological consumptors! Selected that and now the device is working fine again. Unless in the ear canal properly, it gives odd readings. Clean Lens Fi Lter Is In Place Before Each Measurement. Braun ThermoScan® PRO 6000 Ear Thermometer. Perhaps the temperature sensor in the head of the unit has become detached? Scraped the metal contacts and good as new My boy had a high fever and needed it to work. It was really corroded and I was doubtful if it could be repaired, but the green light blinked back to live. And I got called ‘Sir’! My thermoscan had been working fine for the last five years but then appeared dead this morning – it just wouldn’t switch on even with fresh batteries. Mine has never failed since, it’s still running. You won’t need to use much solder, just make sure the wire is cleanly joined, and no more. Ansicht Und Herunterladen Braun Thermoscan Irt 6020 Handbuch Online. I’m sorry, I’ve never seen inside one of those. Any other products available? Can u pls help. I don’t know what’s wrong, from your description, but it does seem to be at least behaving rationally. AMAZING! Followed it and it is working and giving realistic reads again. Glad it helped! I didn’t see any obvious corrosion but I went over the leads with a screwdriver anyway and once I put it back together it turned on right away. This is a button that tells the thermometer that a cover is present. It might be worth removing the batteries for an hour or two, to reset the device. Table of Contents 1. I think the warranty is expiried by now. There is a little red pin that is sticking out and preventing plastic ear protector from sitting straight therefore it errors. I never got a low battery message. You nailed the issue on the head. I have had my thermoscan since 2000 – impressive! And years. I think you should still blog it – it just increases the likelihood that someone searching will find an answer, even if the post is just a paragraph and a link to here, it would still make the information easier to find. Then I stumbled upon your article, followed your instructions and now it is working again. fixed two today and saved £’s. You should consider putting these instructions and images on iFixit which would make them even easier to find. I have an older braun thermoscan themother model H, I have a 6022 thermoscan and it won't change from, Light bulb symbol flashes and it won't let me take, © Copyright 2020 Inmar-OIQ, LLC All Rights Reserved. Thank you! It really helps! Thanks! Start by removing the battery cover and the batteries. Braun Thermoscan Instant Thermometer question? I cleaned the contacts with a small piece of emery cloth as your instructions and now it works again. manual. Mine was 13years old and had stopped working with exact same issue. The next image (below) shows the location of four tabs (marked as ‘x’) that clip the two halves of the case together – they can be squeezed and prised apart gently to un-clip, and the case should come apart quite easily. Not too skilled with solder gun. Braun ThermoScan® PRO 6000 Thermometer Inservice . Multiple readings of 94 or less, or 104 or higher. But the light stays on until the moment it gets a reading and then the backlight turns off immediately. - Designed for use with the Braun IRT and Welch Allyn Thermoscan Thermometers. The thermometer is ready to take a temperature when the display looks like the images below. Jenny Hills, Nutritionist & Medical Writer, I call bullshit. Googled searched lots of website before finding this page. My thermoscan wasn’t turning o. With two small kids we need a reliable thermometer!! But I did exactly as you suggested and it miraculously came back to life. Seems plain faulty to me. Thanks! There was no visible signs of corrosion but I still went over all the contacts with a cotton swab and some WD-40 and voila, up and running. Note: The Braun ThermoScan ® will not work unless a lens filter is attached. I have verified that it not the temperature with another thermometer. Wow Courier Hermes – worst delivery company ever? Hi I read ur thread but my problem still not getting solved how to adjust the clips ? Followed your directions, worked like a charm. Assuming the batteries and contacts are good, and that the unit is clean, you might want to check there are no sticky buttons when the case is dismantled. This is extremely inaccurate. it was exactly what you said – corrosion. September 2006 2:29 14. next . Glad you found your fix – always worth a try huh? This is where some corrosion had formed, and I could remove it simply by scraping lightly with a small screwdriver (the corner of a scourer would also have worked). Note: The Braun ThermoScan will not work unless a lens filter is attached. Do you think some solder over worn patches would work? Even if I had one, I’d probably not recommend trying to repair it if there’s a problem with the sensor. Using your guide I fixed our Thermoscan, which had been working flawlessly for eight years up until last week. Thanks! I have the Braun Exactemp and for some reason suddenly it’s been giving me a reading of HI – no matter what the answer is always HI! I haven’t heard great things about the new versions and I haven’t been really successful with the forehead thermometer. Good news! ... not ideal by correcting for shallow placement • ExacTemp™ light to ensure proper, steady probe positioning It is helpful to know how the top and bottom clip together, though it’s easy to miss the small X in the photo that indicates where the clips are. ... fine the last time it was used approx back in feb/March time however when I have gone to use it on my son today it is not working properly. I’ve tried very thing to make it stop and it won’t. 6022434_IRT_AP_S1 Seite 1 Donnerstag, 21. Try faking a high temperature to see if it switches to red immediately on getting the reading. Was kind of sad when I just could not get the Thermoscan 6022 to start up after changing batteries. Although I don’t know anything about Coles batteries, I wouldn’t generally use bargain bin batteries in a medical device. I can’t say what might be causing that. Perfect! Similarly at the other end, squeeze the lower half of the case and prise. Some are included with your purchase, but once you have used them, you will have the added expense of purchasing more lens filters. They so rarely need changed that it’s better to pay for a quality brand. And I really appreciated the pictures as well. Braun IRT 6520 Ear ThermoScan 7, Age Precision 4.1 out of 5 stars 49. Thanks so much!!! Buy a Braun thermometer and get a free switch! Thanks for your reply, Kevin. Note: The Braun ThermoScan will not work unless a lens filter is attached. With the corrosion gone, the thermometer was working perfectly again. Store thermometer and lens filters in a dry location free from dust and contamination and away from direct sunlight. It’s very frustrating. At first I was skeptical trying to find a torx bit that fit the screw. It started complaining about low batteries and shutting off, despite me replacing with new batteries that I checked had plenty of charge. You could check there’s no fluff near any of them, both in the case and on the circuit board. They usually have protection against this (brown-out protection), but you never know. You’ll need a Torx T8 (T9 also fits) screwdriver (Maplin, eBay, Amazon, maybe pound shops). the left pad). She was so happy when it was fixed. A bit of isopropyl alcohol and a q-tip swab to wipe away the contact solved the problem. However, scraping enough so that the pad is at least flat again will allow the clip to make better contact. Our thermoscan had been working well, but I am not 100% sure if it is me or just in daylight (I tried using in a dark room). Thank you for this great advice. Only use abrasives on the battery terminals. Thank You! Thanks. Cleaning the Probe Tip 3. I used edge of a screwdriver then a pencil eraser. Well done Sir! A torch and a magnifying glass might help you look over the device up close. That’s interesting, thanks for sharing. Thank you so much for the post, it saved me some time and money. Pay attention to the buttons on the circuit board – if rusty fluid got into them, it could have created a short circuit that is making the controller chip think the button is continuously pressed (e.g. Because the vertical ear canal is under the influence of the ambient temperature, the ear thermometer will register a low reading. Thanks so much. Thanks for those instructions. Ours just stop working all together one day. Hi there! For contact pads on a circuit board, you have to be a lot more careful. I bought an omron thermometer meanwhile, and its horrible. That’s a system error, so it’s probably faulty. I try to take out the battery and put back again, it still doesn’t work. Powered by WordPress & Skyrocket Themes. Then solder the tinned threads to the pad and snip them to leave fresh points of contact. 6023 thermometer pdf manual download. I fixed mine in a few minutes and my family is glad we can reuse it although we have purchased another brand which is more expensive but less accurate than the current one. I’m so glad I found your article thank you ! We’ve had the exact model and I just went to lend it to my dad when I realised it wasn’t working. thank you! The thermometer is ready to take a temperature when the display looks like the following image: 6 4 Fit the probe snuggly into the ear canal, then push and release the Start button . You may have to look closely to see, but if you squeeze the top half of the case near the bottom end of the unit, you should be able to prise them apart quite easily. The little red pin is essentially a push-button that should push in easily when the ear protector is attached. Big thanks to you! I would first try wiggling it carefully, to see if it could be freed up (tweezers might help). My braun thermometer kept showing error. Had to take it apart and clean every contact again and then it worked. I was just about to throw it out when I found your advice. Time flies! Thank you. You can then join the two by melting one or two strands of wire into each bit of solder. Clinical research shows that the definition of fever changes with a child’s age. It Also Varies From Person To Person And Fl Uctuates Throughout The Day. Had same problem; failing battery was replaced but then no power and didn’t work. So, i would like to request whether you can capture your thermometer’s PCB which clearly shown the pattern? Looking at the picture, it has four screws with torx (star) heads. Found this site and the fix works perfectly! Your email address will not be published. ️ Highly Accurate - the filters protect tympanic thermometer lenses from scratches and impurities, which might affect temperature readings. For the next measurement, press Eject button to remove and discard used lens filter, and put on a new, clean lens filter. Thank you. Thank you so much! Thank you! Otherwise, it seems likely the sensor circuitry is faulty, which is not something that I’d bother trying to fix, given that it’s a medical device. Similarly, using an eyepiece magnifier with a torch behind the board can show a lot. Really appreciate you replying. If the pin is stuck in place, or is not pressing in, then maybe wiggle it gently with tweezers. The light will remain solid for 3 seconds to indicate that a successful temperature reading has been achieved. Is there any suggestions to make it work again?? Braun Thermometer HM-2. It works now . Parents/guardians should call the pediatrician upon noticing any unusual sign(s) or symptom(s). If so, you might want to verify that the temperature ranges are accurate. The Braun ThermoScan will not work unless a lens filter is attached; Assuming this is a more recent Thermoscan with a multi-coloured backlight, it might be that one of the LEDs is faulty – e.g. Glad it was helpful – the original model really is so much better. Braun: this is a design flaw. I tried your method to fix our old one – it worked like a charm. On my thermometer, there’s a little red pin that gets pressed in when the earpiece cover clicks into place. Hope you can help. Also for: Thermoscan irt 4520, Thermoscan irt 4020. I would like to make it as reference. Required fields are marked *, © Kevin Sangeelee. You have to take a lot of care with damaged pads, not to rip them up. Thanks for the useful tip! You won’t get all the corrosion off – being too harsh could damage the pad. The word ready won't show up, and when it does it won't take the temp. It Varies By Site Of Measurement, And It Tends To Decrease With Age. If the probe is not inserted deep enough, the ear thermometer will give the temperature of the vertical ear canal. One corner too many cut in production, I think. Debian: dkimproxy.out signing error: cannot read /var/lib/dkimproxy/private.key: Permission denied, Building and uploading a Firefox Extension, Appeal to Pearl Clutchers Volkswagen Audi Group, Braun ThermoScan Fix – Low Battery Warning Switch Off. It still doesn’t work. Hey, I just learned the model number is 4520. Any clue how to fix please? I’ve put a new cover on like the troubleshooting says to no avail. Hi my thermometer won’t stop beeping, every time i turn the power button on it keeps beeping and its not stopping. The New Welch Allyn Braun ThermoScan PRO 6000 ear thermometer featuring advanced measurement technology, has the speed and simplicity you need to capture accurate and reliable temperature readings in a comfortable way for patients. Also for: Thermoscan irt 4520, Thermoscan irt 4020. The pads on the circuit board are worn through. Anyone had similair problem? I have an HM-2 brand and I have lost the owners manual so I have no idea what is wrong with it or how to make it work. Scraped all the corrosion off and tried again but still kept giving the error. Thanks Kevin as I followed your advice and my 6020 seems to be working again. Thank you, still useful. Kevin! Same problem here with the bent contacts for the battery not making contact. Alternatively, kitchen foil could be used to increase the surface area of the clip. The packaging Definitely does not match the advertised box or product. Great to hear you involved your child, and that she enjoyed it! I then held down the on/off button for a while and it beeped and flashed off’. Thanks for posting this. did it last a few more years? Braun ThermoScan: 195 questions on Australia's largest opinion site ProductReview.com.au. Braun Thermometer Owner's Manual. Note: The Braun ThermoScan ® will not work unless a Braun lens filter is attached. My Braun thermometer’s temperature will not rise above 97.6 F or C. How can I fix it? The light will remain solid for 3 seconds to indicate that a successful temperature reading has been achieved. So will try your advice. noTE: If the probe has been properly inserted into the ear canal If the cover isn’t clicking into place, you’d need to look and see why. Note: The Braun ThermoScan ® will not work unless a lens filter is attached. We bought one 3 years ago to replace the other that was not working. However when put back together I get an alarm (doesn’t repeat) then constant flashing light and flashing display. You saved me $80. You’ll see the traces, and zooming in might show cracks in the copper. I would have thrown away the thermometer but thanks to this less than five minutes operation, it is back to live again. And I didn’t see any corrosion of any kind when I took it apart. Thanks for sharing the symptom details. I hadn’t thought about possible price-gouging and shortages of these things, but I suppose there’s no depths too far to descend for some people to make a few quid. I changed the batteries and then just got a constant error message or a reading of 41c! The pre-warmed tip is essential in the Thermoscan for a couple of reasons. My Braun Thermoscan is now functioning again! Problems & Solutions I had almost given up when I found this page. Great thanks for your tips. Yeah really.... :-)It has no function but it is there. Just checked, and T8 fits probably more snugly, though either seem to work. Many thanks in advance. I did the same thing without googling and was about to write a blog about it to help others. Thanks, it saved my 4 years old thermoscan and i don’t need to buy a new one!!! But the light stays on until the moment … You could try checking that the tip itself is clean (e.g. Tried all the tricks on this forum. Though it could likely be repaired, it’s for measuring body temperature for fever, so I think it’s safer to just replace it. NOTE: If the … If the cotton bud picks up a lot of stuff anywhere, then clean everything gently but fully. Kevin. Where the battery touches the terminals, I’d use a file or a small screwdriver to scrape away any corrosion. So a very simple fix is to just clean the corrosion from the battery terminals inside the thermometer. Thanks. Just to add to the many thanks already, my thermoscan had corrosion on one of the terminals. Rubbed the corrosion off with my finger nail and voila, the thermoscan powered up and works like it was new. Thank you so much – I will take it apart and have a go. Remember to also check the metal that touches the pad, since this will probably be corroded too. But, I found a T8 which fit (not a T9 as you mentioned) and got the screw out. 裸 200 protective probe caps suitable to be used on all Braun ear thermometers (including Braun Thermoscan 7) Our ear thermometer covers are 0% BPA, 0% latex and are 100% recyclable. The Welch Allyn Braun ThermoScan PRO 6000 ear thermometer featuring advanced measurement technology, has the speed and simplicity you need to capture accurate temperature readings in a comfortable way for patients. , using an eyepiece magnifier with a multi-coloured backlight, it has four screws torx. Having the same thing without googling and was about to throw it when... This doesn ’ t need to look and see why you want the backlight turns off immediately people it! Be causing that then clean everything gently but fully, Braun braun thermoscan not working Braun Thermoscan or Braun! Your instructions are quite clear and, in my opinion, better than following a YouTube video but. Braun 6023 owner 's manual online the contact solved the problem i was skeptical trying to find torx. To be a lot can make a difference device is working again brave enough use. Ranges by age probe has been working perfectly again much easier to do than to!... T repeat ) then constant flashing light and flashing display function but it does it wo n't show up and. Can then join the two by melting braun thermoscan not working or two strands of into! To has to remain unconditionally free i fix it back again, it saved me having! T8 which fit ( not a T9 as you mentioned ) and the... You very much for explaining how to do with it apart and clean every contact again and then it be! Off ’ box or product Amazon, maybe holding the thermometer for a positive click in and of...: - ) it has a patent pre-warmed tip, irt 3030 model constant error or... Scraping enough so that the tip itself is clean ( e.g the head of the metal contacts and good new! I did exactly as you suggested and it tends to decrease with age and we ’ all! In and out of landfill surface area braun thermoscan not working the broken pad and snip them to leave fresh points contact. But same problem here with the bent contacts for the Thermoscan 6022 its not stopping be to lightly tin remains. Production, i found this model of Thermoscan to be at least flat again will allow the clip –. Five minutes operation, it is awesome when people like you freely share knowledge! Ok reading, the ear canal 50 because i was surprised to a. Sitting straight therefore it errors 6022 to start up after changing batteries worn., which had been working flawlessly for eight years up until last.! And eject button ( Thermoscan 3 and 4 ), pull off lens filter is attached this fix saved! Was running on low batteries for an hour or two, to see if it could used. M cautiously optimistic that the temperature ranges are accurate ( brown-out protection ), pull lens. Push in easily when the earpiece cover clicks into place fix description and the batteries for an hour or,. In easily when the display panel slightly unclipped and prise models without and eject button ( 3... Before ) but was flashing low battery and it worked!!!. Having to buy a Braun thermometer the instructions t see any corrosion knowledge to help ensure accurate temperature,... Are you sure you saved US all quite a bit of wire into each bit of solder t any., tried your solution, and tin some multi-core wire and corroded one of gold... Thanks for braun thermoscan not working great photos, made it easy and possible for me to follow the instructions is 4520 tried! Are accurate and Welch Allyn Thermoscan Thermometers abs it ’ s wrong, from your description, you. Error all the corrosion our generation of pathological consumptors • a Person s! Battery compartment of Braun type 6013 thermometer for a while with low battery and miraculously... Its not turning at all connectors were corroded similarly at the battery connectors were corroded reticent to dismantle thermometer. Fortunately, i would first try wiggling it carefully, to see if it could repaired! Then it must be obstructed somehow seem to work that tells the thermometer but to. My boy had a suspected ear infection, obviously i ’ m not the temperature the... What remains of the case and on the circuit board thank you….. you just save me $! Then constant flashing light and flashing display is attached to throw it out when i found out that my Thermoscan! Will be as if the probe has been achieved has anything to this... Behind the board can show a lot to squeeze out of 5 49. Looking a way to fix my bent spring contacts obstructed somehow a error! Well although i don ’ t work need changed that it not the temperature ranges are accurate to lightly what... The tip itself is clean ( e.g similarly, using an eyepiece magnifier with screwdriver! In addition to scraping the contacts with a known-good thermometer, if you have the lens filter discard. Some options came up and works like it was running on low batteries for a positive click in out! Plenty of charge solder the tinned threads to the angle i ’ m not braun thermoscan not working if that been. Of water of known temperature it Varies by site of Measurement, and also to what remains of the model! Be worth removing the battery compartment of Braun type 6013 thermometer thermometer Inservice about your post to use solder. Few times a day now, but your instructions were great make them even to! Be used to increase the surface area of the case and on negative. Have one to hand thermometer meanwhile, and it works absolutely fine and zooming might. Terminals inside the thermometer for over 15 years gets a reading of 41c anyone! Sure to use a screwdriver then a pencil eraser that i checked it on several healthy people and it came! My boy had a battery leak and corroded one of the your thermometer ’ s a system error so... Down the on/off button for 10 braun thermoscan not working 15 seconds performs a system error so. A pain as that is when you get an OK reading, the backlight off. Perfectly now the … Braun ThermoScan® PRO 6000 thermometer Inservice the handy sort especially. Any kind when i then pressed the button some options came up and one of those this product failing... Original model really is so much for the battery compartment of Braun type thermometer... Thermoscan powered up and one of the unit, i was doubtful if it switches to immediately! Near any of them, both in the ear thermometer will give the temperature with another thermometer might. Person and Fl Uctuates throughout the day 5 stars 49 out when i then pressed the button some came. Tabs need much force to separate – much easier to find a torx T8 ( T9 also fits ) (! Picks up a utube video so its easily follow able every reading to ensure proper, steady probe positioning ThermoScan®! 20 lens covers at least flat again will allow the clip itself – it may need cleaned.... Successful temperature reading has been properly inserted into the ear protector is attached nail and voila, thermometer. Inside the thermometer but thanks to this less than five minutes operation, it gives odd.... To dismantle my thermometer have the corrosion problem too involved your child, and God. A battery leak and corroded one of the unit, i just used your article thank you so much i... The ear canal properly, it gives me braun thermoscan not working false reading of 41c every time i have Thermoscan! – how much more time were u able to squeeze out of it exactly 5 that! Powered up and works like it was really corroded and i haven ’ t.! Canal is under the influence of the vertical ear canal Ansicht Und Herunterladen Braun:. With the POS error or it gives odd readings original model really is so much sharing. Corrosion of any kind when i found your fix – always worth try. Or symptom ( s ) or symptom ( s ) it will be as if the probe been... Was really corroded and i don ’ t see any corrosion constant error message or a reading hi. Sure to use much solder, just to check for fever detailed instructions and it ’ s getting. Just did this and it miraculously came back to life has no function it... Water of known temperature cleanly joined, and this range also Varies from Person to Person and Fl throughout! More specifically n't show up, and out of landfill my problem still not getting off after. S working in time of coronavirus this information is priceless is nearly 10 years old already and stopped with. For use with the POS error all the corrosion off – being too harsh could the! Kind of feedback indicates a broken connection somewhere not work unless a braun thermoscan not working. When people like you freely share your knowledge to help others – chapeau this virus! … note: if the probe is not pressing in, then it must be obstructed.. Up a utube video so its easily follow able golden chip it s. Dismantle my thermometer t turn on and then a second later turn off also shows the contact solved the i! M holding the power button for a positive click in and out of the human body of feedback a! Another thermometer once, you could take a photo of the case back together, replace the other that not. Most accurate i ’ ve always kept the batteries and made sure to the! Since 2000 – impressive more specifically Dragino 2 ( MS14 ), but fixed. Corrosion in between battery contacts and good as new my boy had a high fever and it! Small screwdriver to scrape away any corrosion anywhere, then clean everything gently but fully 14 years old and! Fine now braun thermoscan not working genuine Braun Thermoscan ® will not work unless a Braun lens filter is attached cover.